Atmospheric Pressure: 50 kPa to 106 kPa
ALWAYS power off the unit and unplug the AC / DC power cords from the outlet and the unit before cleaning. DISCONNECT the pad from the unit.
The unit can be cleaned with a neutral disinfectant wipe. Pads can be cleaned with a neutral disinfectant wipe, soap-water, or mild detergent. Then, rinse pads with warm tap water and dry with a clean, lint-free cloth.
When used in accordance with the cleaning & disinfectant wipe manufacturer’s instructions for use, the following cleaning & disinfection wipes do not adversely affect the appearance or performance of the ThermaZone Unit or Pads:
Clean Unit anytime it becomes dirty or between uses with different patients. Clean Reusable Pad anytime it becomes dirty or between uses.
FLUSHING THE THERMAZONE SYSTEM:
Under normal use, the internal water can become contaminated with dust and water precipitant. This is a normal occurrence and periodic flushing of the system may help alleviate potential clogging of the water reservoir and internal plumbing system. Flushing of the system once a month is adequate, but heavy use may require more frequent flushing. Please see the “Preventive Maintenance” section of the Instruction and Safety Manual for complete flushing instructions.